Payment & Shipping Terms:
|Weight:||330g||Size:||148 X 60 X 23 Mm|
|Warranty:||12 Months||Suitable For:||Mindray IMEC8, IMEC10, IMEC12, IPM8, IPM10, IPM12|
rechargeable battery pack,
rechargeable battery set
11.1V 4500mAh Li-ion Monitor Battery For Mindray IMEC8 IMEC10 IMEC12 IPM8 IPM10 IPM12 Monitor -- LI13I001A
|Size||148 x 60 x 23 mm|
|Suitable For||Mindray IMEC8, IMEC10, IMEC12, IPM8, IPM10, IPM12|
Detail Image & Replacement Display
1. If the patient monitor is installed with a battery, remove this battery first.
2. Verify that the patient monitor works correctly when running powered form an AC source.
3. Insert this battery per the procedures provided in the Operator’s Manual.
4. Remove the AC power cord and verify that the patient monitor still works correctly.
1. When first installed.
2. Whenever a battery is replaced.
The monitor can operates correctly from battery power when an AC power failure accidentally occurs.
Perform the test by referring to the Battery chapter in the Operator’s Manual and verify the operating time of the battery meets the product specification.
Once every six months or if the battery run time reduced significantly.
The patient monitor can operate independently on a single battery.
Battery Use Guidance
1. The lifetime of the battery is about 3 years.
2. A battery performance check every two years.
3. Optimize the battery once every 3 months.
4. When the monitor will not be used for a long time, the lithium battery should be taken out and preserved.
5. the battery should be replaced immediately when the battery is damaged.
Battery Optimization Steps
1. Battery charge over 10 hours.
2. Disconnect the alternating current, use the battery power until the monitor closes.
3. Recharge the battery for more than 10 hours.
Battery performance check: the first two steps to optimize the battery, and the battery power time reacts to the battery performance.
Battery life: a new full power lithium battery, the environment temperature is 25 DEG, the typical configuration (continuous pulse probe, ECG cable; Temp cable; NIBP work in the time interval for the 15min automatic measurement mode) under the condition of continuous working time is not less than 4h.
1. Discharge the lithium battery to 50% and remove it.
2. After six months of storage, discharge the lithium battery and then full charge.
3. Repeat the above steps.
This battery has been manufactured with high-quality components. It complies with every particular of the orginal manufacturer's regulations and is guaranteed against all manufacturing defects.
A margin of 0.1V difference in the voltage is negligible in batteries. Therefore, such a battery would still be perfectly compatible. If the capacity (mAh) of the replacement battery is higher than that of the original battery, this will allow for a longer usage time.
Big wholesale order is acceptable and the price is negotiable, please contact us for details.
OEM is also welcome. Please tell us the details of the product you want, then we can design and customize it for you.
Do not hesitate to contact us anytime before and after your purchase. If you have any questions on your purchase, please let me know via "contact now" or by email. We will solve it for you.
We are committed to ensuring that you are completely satisfied with every transaction. 100% Satisfaction guaranteed !
We will reply your mail within 24 hours normally. sometimes have a little delay due to the weekends or holidays.
(we work from Monday to Friday (8-17:30 GTM+8)
If you can't get our mail after 48 hours, please check your spam or contact to us. If you email us and I will be on service of you until you are satisfied with your purchase.
Contact Person: liu