Payment & Shipping Terms:
|Weight:||96G||Size:||67 X 37 X 20mm|
|Warranty:||12 Months||Suitable For:||BIS Vista Monitoring System, BIS VISTA, VTI 14564|
7.4V 2600mAh Li-ion Monitor Battery For BIS Vista Monitoring System BIS VISTA VTI 14564
|Size||67 x 37 x 20mm|
|Replace Code||185-0152, 186-0208|
|Suitable For||BIS Vista Monitoring System, BIS VISTA, VTI 14564|
This battery has been manufactured with high-quality components. It complies with every particular of the orginal manufacturer's regulations and is guaranteed against all manufacturing defects.
A margin of 0.1V difference in the voltage is negligible in batteries. Therefore, such a battery would still be perfectly compatible. If the capacity (mAh) of the replacement battery is higher than that of the original battery, this will allow for a longer usage time.
In the event of a power failure or interruption of power during a procedure, the monitor automatically switches to back-up battery operation. A fully charged battery will provide approximately 45 minutes of operation. When the system is running on battery, a battery icon displays, indicating the battery status. When the battery reaches a low power condition, the monitor beeps and the battery symbol displayed on the screen changes color from green to orange. In addition, a “Battery Power Low” message blinks continuously in the Message area of the screen. The Save Settings feature is disabled when the battery power is low. Battery recharge time is approximately 6 hours.
The battery must be tested periodically to verify that the machine will continue to operate during power outages. To test:
1. Charge the machine by leaving it plugged into A/C power for at least 6 hours. The monitor charges while in either Standby mode (yellow light) or ON (green light).
2. Disconnect the A/C cord from the wall supply.
3. Verify that the machine operates reliably for 45 minutes.
4. Recharge the battery.
ELECTRICAL SHOCK HAZARD: DO NOT REMOVE MONITOR COVERS DURING OPERATION OR WHILE POWER IS CONNECTED TO MONITOR.
GROUND WIRE LEAKAGE CURRENT MUST BE CHECKED WHENEVER INSTRUMENT CASE IS OPENED BY A QUALIFIED BIOMEDICAL ENGINEERING TECHNICIAN.
Check the battery periodically by operating the monitor that has been disconnected from the wall socket and that has been charged to full capacity (at least 6 hours of charge time). After long periods of storage (e.g., more than 1 month) it may be necessary to cycle (charge, then discharge) the battery a few times to get full charge capacity. If the monitor fails to operate reliably from the battery for approximately 45 minutes, battery replacement is required.
The monitor contains an internal lithium ion battery. The battery must be removed by a qualified service technician and disposed of or recycled in accordance with the national laws of the country.
Big wholesale order is acceptable and the price is negotiable, please contact us for details.
OEM is also welcome. Please tell us the details of the product you want, then we can design and customize it for you.
Do not hesitate to contact us anytime before and after your purchase. If you have any questions on your purchase, please let me know via "contact now" or by email. We will solve it for you.
We are committed to ensuring that you are completely satisfied with every transaction. 100% Satisfaction guaranteed !
We will reply your mail within 24 hours normally. sometimes have a little delay due to the weekends or holidays.
(we work from Monday to Friday (8-17:30 GTM+8)
If you can't get our mail after 48 hours, please check your spam or contact to us. If you email us and I will be on service of you until you are satisfied with your purchase.
Contact Person: liu